Utilize este identificador para referenciar este registo:
|Título:||Viscosity measurements of three ionic liquids using the vibrating wire technique|
|Autor:||Diogo, João C. F.|
Caetano, Fernando J. P.
Fareleira, João M. N. A.
Wakeham, William A.
Reference ionic liquid
1-Ethyl-3-methylimidazolium ethyl sulfate([C2mim][EtSO4])
1-Ethyl-3-methylpyridinium ethyl sulfate([C2mpy][EtSO4])
|Resumo:||The paper reports measurements of the viscosity of the ionic liquids 1-hexyl-3-methylimidazoliumbis[(trifluoromethyl)sulfonyl]imide ([C6mim][NTf2]), 1-ethyl-3-methylimidazolium ethyl sulfate([C2mim][EtSO4]) and 1-ethyl-3-methylpyridinium ethyl sulfate [C2mpy][EtSO4], by means of avibrating wire viscometer. Density measurements on the same samples have been carried out usingan Anton Paar U-tube density meter. Recently, we have studied the applicability of the vibrating-wire technique to the measurement of the viscosity of ionic liquids. The fact that these liquidsare electrical conductors represents a source of additional experimental uncertainty, if a baremetallic wire is used as a sensor. The previous study involved the measurement of the viscosityof trihexyl(tetradecyl)phosphoniumdicyanamide [P6,6,6,14][dca], and it was shown that the effectof the electrical conductivity was essentially insignificant. The current study extends our work toliquids 1-hexyl-3-methylimidazolium bis[(trifluoromethyl)sulfonyl]imide ([C6mim][NTf2]), 1-ethyl-3-methylimidazolium ethyl sulfate ([C2mim][EtSO4]) and 1-ethyl-3-methylpyridinium ethyl sulfate([C2mpy][EtSO4]) which have an order of magnitude higher electrical conductivity than [P6,6,6,14][dca].The main aim of the present work remains the study of the effect of the electrical conductivity on theperformance of the vibrating wire instrument used. One important difference from our previous work,however, is the fact that [C6mim][NTf2] has been proposed to be a reference fluid for several thermo-physical properties by IUPAC, for which a round-robin set of thermophysical property measurements hasrecently been carried out. Therefore another of the main goals of this work is to compare our vibratingwire results for the viscosity of the reference liquid [C6mim][NTf2] with those data. The present resultsagree within ±2% with a correlation of those results obtained with the IUPAC sample that report watercontamination below 120 mg kg−1.|
|Versão do Editor:||http://www.sciencedirect.com/science/article/pii/S0378381213002252|
|Aparece nas colecções:||Ciências e Tecnologia - Artigos em revistas internacionais / Papers in international journals|
Ficheiros deste registo:
|2013_viscosity_measur_of_3_ILs_using_VW_technique_Fluid Phase Equilibria.pdf||905,61 kB||Adobe PDF||Ver/Abrir Acesso Restrito. Solicitar cópia ao autor!|
Todos os registos no repositório estão protegidos por leis de copyright, com todos os direitos reservados.